- Total Chlorine Ultra Low Range Colorimeter HI761 Hanna Checker - Fresh Water
Four simple steps to results. Easily and accurately test for ultra-low range total chlorine in with the Hanna Checker Colorimeter - Total Chlorine Ultra-Low Range (HI 761). Designed to be more a more accurate alternative to chemical test kits. First, add sample to the included cuvette. Next, insert the sample into the Checker and zero the instrument with your water sample. Then, remove the sample and add the reagent. Finally, place the vial into back into the Hanna Checker, press the button and read the results. Done.
Easier to use and more accurate than chemical test kits.
The HI 761 from Hanna Instruments uses an adaptation of the USEPA method 330.5. The reaction between the chlorine and DPD reagent causes a pink tint in the sample. Auto shut off after 10 minutes of non-use.
Dedicated to a single parameter
Two sample cuvettes and caps. Six packets of reagents for total chlorine ultra-low range
Small Size, Big Convenience
Weighing just 64 g (2.25 oz.), the Checker®HC easily fits into the palm of your hand or pocket. Use for quick and accurate on the spot analysis. Single button operation: Zero and Measure Operated by a single AAA battery.
- Range:0 to 500 ppb
- Resolution: 1 ppb
- Measurement Method: Colormetric
- Accuracy: ±5 ppb ±5% of reading
- Light Source:Photodiode @525 nm
- Light Detector: Silicon photocell
Used on RO water By Anthony on 3/1/2018Anthony would recommend this product to a friendChecked before and after my carbon blocks, max on before and zero after on 6 months plus carbon blocks.
one of the most important testers there is By Lance on 3/1/2018Lance would recommend this product to a friendI say that b/c chlorine/chloramine kept my corals from living for many years. Using this I know when my chloramine removal carts need changing. Light pink on the colormetric tests can look like clear, and you can be ending up with chloramine and chlorine in your tank without knowing it. Using this along with chloramine monster and dual chloramine pentek carts after I'm secure.
nice checker By moses on 5/31/2014moses would recommend this product to a friendIt seems to work really good I just wish you could use a liquid instead of the powder reagent < its a pain to get it all out. It shows that the carbon filters that I bought for chloramines from BRS takes most but not all out of the system still leaving 40 ppb.