Spot on! By Christopher on 11/12/13Christopher would recommend this product to a friendThis is a great conductivity probe. Some have reported problems and I to had problems initially. I found for me the key was to be patient when the probe is in the settling process during calibration. It takes some time for the probe to settle out. Another issue was placement. I have this probe in the return section of my sump where circulation is great and micro bubbles are at a minimum. One last thing is the temp probe, you must use a temp probe to gain accurate and STABLE readings. If not your salinity reading will change with temperature. Great product by neptune and great service and prices from BRS!!!!
Had some difficulty... By Russ on 10/1/13Russ would recommend this product to a friendI couldn't get mine to calibrate properly through the manual interface and had to do it on the web interface, even after updating firmware. Reading about it on forums says I wasn't the only one with trouble and a lot of people had trouble getting it to calibrate accurately. Still, it seems people (as well as myself) atleast got it really close so we can use it to monitor for changes which is all you really need. Knowing what I know now I still would have bought it, but I think they could have specified a little more that salinity isn't really exactly the same as conductivity.
Excellent product, accurate & dependable By Vanore on 9/13/13Vanore would recommend this product to a friendI was put off on this probe for 6+ months due to the price. I have had it running for less than a week now, and I seriously should have purchased it with my original apex order. If you own a hydrometer, forget about it. use it for backup. This probe calibrated fast, and is spot on. It is an incomparable tool once you use it. BRS also had the best price, and shipping rates.
View Larger Map 6325 Sandburg Road, Suite 400, Golden Valley, MN 55427