Okay By sh55 on 3/10/14sh55 wouldn't recommend this product to a friendResults are varied, I am not able to get a stable result and when compared to my Salifert Ca kit this kit reads 30-50 points higher.
I recommend watching the video Hanna posted on Youtube for more precise instructions and pointers.
When the instructions tell you to add .1ml of the sample, it is referring to your tank water not the mix in the cuvet or RODI.
Buyer beware not accurate By Rolando on 3/2/14Rolando wouldn't recommend this product to a friendThis product in my option should not be on the market until it is improved, just took three reading of the same water sample first 381 ppm second 449ppm third 495ppm using the same method, I have the Havana alk tester it's great, also the phosphate it's also very good. But don't spend your cash on the calcium tester if you do you'll be sorry you did.
Never the same result twice By Todd on 1/26/14Todd wouldn't recommend this product to a frienddirections are a bit complicated (compared to other Hanna checkers) and it seems that I never get the same result twice. I have gone back to my previous CA test kit.
Good product By Keith on 1/14/14Keith would recommend this product to a friendToo many steps involved for my liking and doesn't always seem accurate but overall I think it's better than most other tests.
7 out of 10 By Jordo on 12/8/13Jordo would recommend this product to a friendThis isn't what you would expect from Hanna. The read out is great to have but the directions are a bit complicated. As a Nurse I am very used to small dose syringes and using precise lab equipment and this is kind of a pain for me. I have tried 3 tests on the same sample of water and got 3 different reading approx 30-40 points off. I'm still getting used to this test so maybe It will get better. I am a little disappointed =(
Improving Ca test precision By Bob on 12/8/13Bob would recommend this product to a friendI purchased recently and like the digital readout but was a little concerned about accuracy and precision, primarily due to difficulty in accurately delivering a very small sample size (0.1 ml). I have made some suggestions to Hanna but in the meantime have revised my test procedure which seems to give me more precise results.
One possible solution would be to use a 0.1 ml syringe, however, these are fairly expensive and fragile. Alternatively, I revised my procedure as follows:
Instead of initially filling the cuvette to 10 ml with RO/DI water, I fill to 8.5 ml after adding 1 ml Reagent A and then press Add C2 on the meter.
To obtain the 0.1 ml sample I add exactly 1.0 ml sample to a clean cuvette and dilute to 10 ml with RO/DI water. I cap and invert 3-5 times. Then I take exactly 1.0 ml of this diluted sample and add it to the test cuvette. Then I finish filling to the 10 ml mark with RO/DI water and then add Reagent B and proceed as usual. This results in a more accurate delivery of 0.1 ml of sample.
The reproducibility and accuracy of the resulting readings are much improved.
Much better with improved reagents By Matt on 10/5/13Matt would recommend this product to a friendI'd been getting variable results, but I found a post that said they had reformulated the reagents to improve the precision. I contacted Hanna and they sent me replacement reagents (and noted that they had recalled the old ones from retailers).
So, I'm getting much better readings now than I was before and if you've written it off, you may want to try it again with new reagents.
Good idea, bad instructions!! By Jason on 4/18/13Jason wouldn't recommend this product to a friendI love the idea of a digital cal. checker.
However, after following the instructions in detail I could not get a good reading.
Had to go to the internet to find out what the problem was.
YOU MUST USE DISTILLED WATER WITH THIS TESTER TO MAKE IT WORK!!!!!!
Checker is inaccurate due to being overly complex and requiring absolutely pure RO/DI By Adam on 4/16/13Adam wouldn't recommend this product to a friendI have had this checker for close to a year now and have finally completely given up on it. It was a waste not only on the checker itself but also on the $20 reagent packs. I have never gotten the same value twice from it. Using the same RO/DI water and sample water. I follow the exact same test procedure every time. My Hanna alk checker gives the same number on repeat tests and is invaluable. But the Ca checker seems to just randomly choose a number. This i believe is due to the 1/10 ratio being used. Perhaps the checker would operate correctly if i used lab grade RO/DI but i am unwilling at this point to spend another cent on it.
Great Product By Mark on 3/9/13Mark would recommend this product to a friendAnything that makes testing easier is for me. I would absolutely recommend this product. I am not good at titration and can never tell when the end color is reached. This is a must for testing.
View Larger Map 6325 Sandburg Road, Suite 400, Golden Valley, MN 55427