Ask a question
Hanna Checker Instructions for Fresh Water Alkalinity Standards:
- 1. Thoroughly wipe standard cuvette with HI731318 lint free cloth and only handle using top of cuvette. The cuvette wall must be dry and free of smudges.
- 2. At the C. 1 prompt, place A, HI775-11 standard cuvette into the holder of the Checker and press operations button.
- 3. At the C. 2 prompt, place B, HI775-11 standard cuvette into the holder of the Checker and press operations button.
- Acceptable results must be between 90 and 110 ppm at 25°C
There are no videos related to this product.